Test Sieve Cleaners
HAVER USC - Ultrasonic Cleaners
HAVER USC test sieve cleaners guarantee that sieves are cleaned thoroughly and gently while at the same time saving energy. They have proved to work outstandingly well in conjunction with the HAVER USC cleaning fluid. USC cleaners are available for cleaning individual sieves and for the simultaneous cleaning of up to five sieves.
Your advantages at a glance
- The sieves are cleaned thoroughly and gently while at the same time saving energy.
- Easy handling.
- Up to five test sieves can be cleaned simultaneously.
|USC 200||USC 500||USC 200 Multi|
|Number of sieves||1||1||5|
|Sieve diameter||≤ 230 mm||≤ 500 mm||≤ 230 mm|
|Oscillation tank (inside) [mm]||Ø 245 x 130||Ø 515 x 200||500 x 300 x 300|
|Charge||5.7 l||42 l||40 l|
|Outlet||Ball valve G 1/4||Ball valve G 1/2||Ball valve G 1/2|
|Outside dimensions [mm]||Ø 260 x 260||Ø 570 x 490||560 x 360 x 500|
|Weight||4.8 kg||22.4 kg||27 kg|
|Operation voltage||230 V or 110 V; 50/60 Hz||230 V or 110 V; 50/60 Hz||220-240 V; 50-60 Hz|
|HF-Frequency||35 kHz||35 kHz||35 kHz|
|HF-constant maximum output||2 x 240 W/per.||2 x 600 W/Per.||2 x 1000 W/per.|
|Power consumption||0.50 A||1.30 A||4.50 A|
|allowable surrounding temperature||5° to 40°|
|allowable relative humidity up to 31°C||80 %|
|allowable relative humidity up to 40°C||50 %|
|defrosting not allowable||x|
Maintenance of HAVER Test Sieves
Test sieves with apertures of 500 µm and smaller should be cleaned in an ultrasonic bath. Bater ultrasonic cleaning rinse the test sieve with clear water and blow out the sieve bottom with compressed air. Let the water drip for about 5 minutes and dry the test sieve at 80° - 100°C max. for about 30 minutes.
Remove any remaining particles from the surface of the mesh by tapping the test sieve frame slightly against a rubber or wooden pad. Never remove particles by brushing.
(see ISO 2591-1, DIN 66165, Part 1 and Part 2)
- Brochure Particle Analysis
(PDF, 1.9 MB)
E-Shop Particle Analysis
With this E-Shop button the USC cleaners and corresponding accessories are reached directly.